Save your pacakage (will need new artwork as well)
Serial Numbers labels
EC Declaration of Conformity
Safety Warning sheet (will need an updated one)
Box part number
Middle Foam part numbers
@DRobertson we will need updated artwork for the following:
Save your packaging- we need this in the blue/black colors instead of LulzBot Green and Black
Updated Safety warning sheet : http://devel.lulzbot.com/mini/hibiscus/production_docs/compliance/safety_warnings_LulzBot_Mini2.pdf
Will also need a Blue Logo to put on the packing and QA sheets: http://devel.lulzbot.com/mini/hibiscus/production_docs/Quality_Assurance_Record_InkJet.odt and http://devel.lulzbot.com/mini/hibiscus/production_docs/Packing_List_Print_Shop.odt
DC-LB0198- Save your packaging
DC-LB0199-Bio NA Serial Number Sticker, Tamper Evident
DC-LB0200-Bio EU Serial Number Sticker, Tamper Evident
DC-LB0201-Bio AU Serial Number Sticker, Tamper Evident
DC-MS0095-EC Declaration of Conformity, Bio
DC-MS0096-Packing List, Bio
DC-MS0097-Safety Warning sheet, Bio
DC-MS0098- Quick Start Guide, Bio
@DaniAO Idk if the glass syringe is considered part of the tool kit even though it doesn't go inside it, but I think we should have a checkbox for it on the Packing List.
Otherwise that one looks good.
E0 steps/mm incorrect, should be: 9448.8
no sure what z endstop offset should be filled with as we wont be doing z offset calibration AFAIK
No spool arm
No PEI sheet
check 5: no tensioning collars on Bio X axis; this can be removed.
remove extruder section entirely and replace with syringe pump
check would be "Does the Syringe Pump Carriage move the full length of travel?"
no LCD knob
no FW checksum
no heatsink fan
no extrusion fan
no nozzle temperature
syringe pump control vs extruder
selecting level x axis should be "during plunger re-home and auto-level routine"
no wipe sequence
no z offset calibration
no bed leveling function
no octo test print (remove yeoldebrian CC by SA comment at bottom also)
I've updated the warning sheet to include sharps. I couldn't find any standardized messaging, so if something is missing, please let me know and I will amend the copy right away. Doc is attached.
On QA record:
Remove far right column of table; no z-offset calibration on Bio
Probably would be a good idea to include something about wiping down the frame if that is something we want to do, but with what, how, and at which stage in the assembly will need to be determined and documented.
selecting level x axis should be "during plunger re-home and auto-level routine
This is because when the machine is powered on, the user is asked if they would like to home, and then prompted again asking if they would like the Syringe Pump moved to the loading position and auto level.
I do not see an option to independently level the x axis through the LCD, so this takes place at machine start up, triggered by those prompts.
So that line would read something like:
Verify that both carriages touch the Z-motor mounts during plunger re-home and leveling routine.
Verify that both carriages touch the Z-motor mounts during start-up leveling routine.
@oliver we need to get the following added to the OHAI/BOM. Can you please let me know which assembly you would like them to go in?
DC-LB0202- Bio certification sticker
DC-LB0203- Bio "Fresh" sticker DC-LB0204- Bio Caution sticker _ this goes along the back of the electronic case that warns customers not to open it